HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||6 November 2018|
|PDF File Size:||14.1 Mb|
|ePub File Size:||2.47 Mb|
|Price:||Free* [*Free Regsitration Required]|
With an external signal source, the HM can also be used in the sweep mode.
HM8030 Function Generator Module
Save to an existing parts list Save to a new parts list. Use a 50 ohms through-termination.
These val- ues are obtained when the generator output is terminated with 50 ohms. This file is downloadable free of charge: Any zeroes preceding the decimal point are dropped. The product detailed below complies with the specifications published by RS Components. Connect multimeter to TP1.
Szeretnem tudni milyen vegyszerrel lehetne tokeletesen kitakaritani. Safety Every module is hamey and tested for use only with the mainframe HM according to lEC Part 1 and la Safety requirements for electronic test and measurement equipment. Therefore only short or low-capacitance cables without a 50 ohms ter- minating resistor should be used for connection. D – 1 kHz square wave 8300 cycle 1. Remove both screws on rear panel and, holding case firmly in place, pull chassis forward out of case.
Please select an existing parts list. If test circuits having a 50 ohms input impedance are con- nected, this termination is not required. Connect the distortion meter to TP1.
B – Triangle amplitude at 1 kHz Setting: A negative DC voltage reduces the frequency. Please, do not make a copy of the downloaded manual and do not offer to sell it, just use personally to repair defective appliance! The frequency changes by a factor of The fol- lowing precautions should be observed: In the open-circuit condition, the available signal amplitude is about twice as high.
The specified values are only obtained if the hameh control knob is turned fully clockwise. Please supply name and department, address and telephone number to ensure rapid service.
Use a 50ohms through-termination.
The required trimmer is located on the upper PCB next to the amplitude control. If the BNC socket at the rear panel of the HM unit was in use before, the BNC cable should be discon- nected from the basic unit hameh safety reasons. HM function generator module,5MHz. The TP 1 test point is located next to the amplitude control on the upper PC board. The Manufacturers and RS disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website.
HAMEG HM Function Generator 10mhz | eBay
Add to a parts list. All outputs are short-circuit-proof.
E – Hz square wave duty cycle Setting: Adjust , until the negative pulse length equals the positive pulse length on the screen.
Save to parts list Save to parts list. We, the Manufacturer or our representatives may use your personal information to contact you to offer support for your design activity and for other related purposes. Trigger output In the sine, square and triangle modes, the trigger out- put 0 supplies a square signal in synchronism with the output signal.
You agree that the Information as provided here by RS nameg not be error-free, accurate or up-to-date and that it is not advice.
Before opening the set, you should carefully read the 80330 and Warranty information on page M2 of these operating instructions. This check should also be carried out for the other signal waveforms. Thank you for your feedback. Set the trigger edge con- trol slope of the oscilloscope to negative edge trigger- ing.
The output impedance is 50 Q, and the max. Furthermore, this cable must be ter- minated with a 50 ohms through-termination e. To open downloaded files you need acrobat reader or similar pdf reader program. Please enter a message. First-time operation of the module Provided that all hints given in the operating instructions of the HM Mainframe were followed – especially for the selection of the correct mains voltage – start of operation consists practically of inserting the module into the right haneg left opening of the mainframe.
Output attenuator function Setting: These free programs can be found on this page: Adjustment procedure The following instructions will permit you to correct any deviation of the module from the nominal data.
The DVM should display 1 V or 0.
HAMEG HM 8030-6 Function Generator 10mhz
When reclosing the module, care should be taken that the guides engage correctly with the front chassis. The logical front-panel layout of the module ensures rapid familiarisation with the various functions.
If con- trol 0 is turned clockwise to the uncalibrated position, the output signal is limited shortly below the upper screen border. Adjust timebase, so that the positive pulse period of the output signal has a width of exactly 10 divisions. General Information Operating condKions: G – Accuracy rating of the frequency ranges Setting: